- B2B Industrial Supplier since 1999
- Shipping from our own warehouse in NL !
- Shipping in EU with DPD Traceable
- Reliable SSL & Secure Payments
The EI-1040 is a dual programmable gain instrumentation amplifier with a precision reference output for bridge excitation. Gains are digitally selected at values of 1, 10, 100, and 1000. Four TTL or CMOS- compatible address lines individually select the amplifier gains.
Applications of this device are for signal conditioning and amplification of low-level signals such as thermocouples and transducers. This device is also used in conditioning signals to be transmitted over a long distance to single ended receivers.
The EI-1040 requires +5 volts DC at a nominal 0.1 amp. An internal DC to DC converter supplies an output of +15 and –15 volts nominal. The EI-1040 consists of 2 Burr-Brown/TI PGA204 amplifiers and one DCP010515DPB DC to DC converter.
A 4.096 volt reference is provided for connection to a bridge or other device requiring excitation. The maximum allowable current draw from this source is 5 ma.
The EI-1040 can be attached to the LabJack by simply connecting the power and amplifier outputs to the LabJack. When connecting the EI-1040 to the LabJack, the LabJack should be powered down prior to making the connection. After the connection is made then the combination can be powered up. The gain of the EI-1040 can be programmed by the LabJack by connecting the gain select inputs GSA1, GSA2, GSB1, and GSB2 to the LabJack digital outputs. The configuration for the gain select is shown below:
|GSA1 or B1
|GSA2 or B2
Combinations of gain can be selected so that for example one amplifier can have a gain of 10 and the other have a gain of 100. The voltage level of a 0 is typically ground and the level of a 1 is 2 volts above the digital ground (see the PG204A spec sheet for exact levels).
It should be noted that when the EI-1040/LabJack is powered from limited power sources such as notebook computers, bus-powered hubs, etc., there may not be enough current to supply both devices. A message from the computer should tell the user of this condition.
The instrumentation amplifiers used are Texas Instruments/ Burr Brown PGA204 parts
Typical applications for this unit include: SIGNAL CONDITIONER, THERMOCOUPLE AMPLIFIER, STRAIN GAUGE AMPLIFIER, DATA ACQUISITION APPLICATIONS, SIGNAL FILTER, AUDIO AMPLIFIER, MICROPHONE AMPLIFIER